Vt-400A Computerized High-Frequency Ventilator Machine Price for Infant
- Model NO.: VT-400A
- CE: Certificated
- Specification: CERTIFICATES: CE, ISO13485, ISO9001
- Type: Ventilator
- Trademark: Foinoe
- Origin: Nanjing, Jiangsu
A. The tidal gas can be exported with constant pressure, contact current and accurately adjustable flow.
B. Opening ventilation, which used opening intubation without casing bag, and get little resistance when expiratory is suitable for the ventilation management of infant.
C. Evident effectiveness of ARDS treatment can be shown with inserting small tube and jetting oxygen through nasal vestibule, which is safe and doesn't cause pressure injury easily.
D. Special CPAP tube for infant ventilation treatment are available.
E. PEEP and humidifier volume can be adjusted accurately.
F. Alarm system with sound and light when no oxygen, power failure or low airway positive pressure.
G. Heated humidifier can be used for good effectiveness of humidifying.
H. Inbuilt storage battery, it can be used while being charged and can be used for more than four hours after full charged.
Main Technique Parameter:
Frequency: 10~150 breaths/min
I/E ratio: 1: 1~ 1: 3 and inverse ratio
Driving Pressure: 0~60KPa
Humidifying rate: ≥ 6ml/h
Process Ventilation: T=10min,
ProI: T1(150bpm): T2(35bpm)=7: 3
ProII: T2(35bpm): T1(150bpm)=7: 3
Storage Battery: 4Ah, 12V, Rechargeable
Alarm: Alarm with sound and light when lack of power, or the airway pressure is overy 1Kpa± 0.2Kpa and alarm will last for 6 seconds when no oxygen.
Oxygen Concentration: 40%-90%(with trolley)
Ventilation Mode: HFJV(High Frequency Jet Ventilation); CPAP(Continuous positive airway pressure); HFJV+PEEP(Positive End-Expiratory Pressure), HFJV+SIGH(Sigh), PRO I and PRO II ventilation. Manual ventilation.
|1.2||Air supply||High pressure oxygen supply (0.2MPa)|
|1.3||Working Power Source||
|2.1.1||Control Ventilation(HFJV)||Adopted mask and atria of nasal or tracheal cannula to supply oxygen.(do not need cuff .)|
It is good for improving the PaO2 of patient during respiratory management in short time.
It is not only good for improving the PaO2 of patient ,but also good for eliminating CO2 during ventilation management in long time.
|2.2||Technical Parameter and Test Parameter|
|2.2.1||Ventilation Frequency Display and adjustment||10~150bpm|
|2.2.4||I/E radio display and adjustment||1:1~1:3 and inverse radio|
|2.2.5||Working pressure display and adjustment||0~60KPa|
|2.2.6||PEEP display and adjustment||0.05~0.2KPa|
|2.2.7||CPAP display and adjustment||0~0.5KPa|
|2.2.8||Airway pressure Display (light beam)||0~1.5KPa|
|2.2.9||Setting of upper-limitation of airway pressure||0.8KPa(settled)|
|2.2.10||Oxygen Concentration||40%~90% ( with trolley )|
|2.3||Inner Storage Battery||4Ah,12V ,rechargeable. It can work for 6 hours.|
|2.4||Alarm||Alarm if no power; alarm after lasting 6 seconds if no oxygen; Alarm for upper-limitation of airway pressure. The function of alarm will be resumed after 2min when you press the muffler.|
|2.5||Weight of Mainframe||7Kg|
|3||Basic configuration and accessories /|
|3.1.1||Mainframe of ventilator||1set|
|3.1.2||Ventilation tube with nasal prongs||1set each|
|3.1.3||CPAP infant catheter||1set|
|3.1.4||Double-head inlet tube (2M)||1pic|
|3.1.5||Pressure measuring tube||1pic|
|3.1.8||Trolley ( contained Arm support )||1set|
|3.1.11||Rubber pneumathode (0.5L)||1set|
|3.2.1||Constant Temperature Humidifier||1set|
|3.2.2||Corrugated tube (1M)||1pic|
|4.1||Technical document||User Manual|
|4.4||Production licence of national medical equipment||yes|
|5||Valid time of quality||Guarantee for 2 years, maintenance in life,but the place of guarantee and maintenance is in China.|